WCPS: Determination of Iodine-129 in Aqueous Environmental Samples by ICP-MS with High Energy Oxygen Collision/Reaction Cell Technique. (PC-330)
Posters | 2011 | Agilent Technologies Instrumentation
Agilent TechnologiesKey words
cps, oxygen, bias, cell, energy, omega, crc, entrance, rnaa, collision, reaction, average, radiochemical, rsd, icp, deflect, gas, accidents, spraychamber, plasma, high, accelerator, deg, power, octopole, nuclear, ions, flow, kinetic, technique, emissions, mass, tuned, exit, accelerate, notice, sufficiently, inductively, plants, separately, newly, reactions, continuously, instrumental, mode, released, units, subject, selecting, using
The ultratrace determination of iodine 129 in aqueous samples using the 7700x ICP-MS with oxygen reaction mode
The ultratrace determination of iodine129 in aqueous samples using the 7700xICP-MS with oxygen reaction modeApplication noteNuclearAuthorsKazumi Nakano, YasuyukiShikamori, Naoki Sugiyama andShinichiro KakutaAgilent TechnologiesTokyo, JapanFirst presented at the 2011 WinterPlasma Conference, Zaragoza,Spain, Jan 30th–Feb 4th 2011.AbstractDetermination of the radionuclide iodine 129...
cps, cpsoxygen, oxygennuclear, nuclearﬂow, ﬂownaa, naaconﬁrm, conﬁrmbackground, backgroundiodine, iodineicp, icpdilution, dilutionoverlap, overlapplasma, plasmafactor, factorkakuta, kakutashinichiro
WCPS: AGILENT 8800 TRIPLE QUADRUPOLE ICP-MS WINTER PLASMA CONFERENCE 2012 POSTERS
AGILENT 8800 TRIPLE QUADRUPOLE ICP-MSWINTER PLASMA CONFERENCE 2012 POSTERSCompendiumLearn more: www.agilent.com/chem/icpqqq4WP08: New Reaction Cell ICP-MS with Improved Ion Optics Arrangement and Vacuum System8ThP09: Efficient Removal of Polyatomic Ions by ICP-MS Equipped with Novel Reaction Cell: Examples of Highly Purified Chemicals...
icp, icpreaction, reactioncell, cellqqq, qqqmode, modeicpicp, icpicpvacuum, vacuumquad, quadcollision, collisionchamber, chamberarrangement, arrangementbec, becion, ionoptics, opticselements
The ultratrace determination of iodine 129 using the Agilent 8800 Triple Quadrupole ICP-MS in MS/MS mode
The ultratrace determination of iodine129 using the Agilent 8800 TripleQuadrupole ICP-MS in MS/MS modeApplication noteNuclearAuthorsYasuyuki Shikamori, KazumiNakano, Naoki Sugiyama, andShinichiro KakutaAgilent TechnologiesTokyo, JapanIntroductionIodine 129 (129I) is a long-lived radionuclide (half-life 15.7 My), which hasbeen released into the environment as a...
cps, cpsabundance, abundancenuclear, nuclearradionuclide, radionuclidesensitivity, sensitivitybeneﬁts, beneﬁtsiodine, iodineicp, icpgas, gasthermalize, thermalizereaction, reactionkakuta, kakutashinichiro, shinichirocell, cellmeasurement
WCPS: Reaction Cell Frontier: Analysis of Radionuclides in Environmental Samples using an Innovative Reaction Cell ICP--MSMS
Reaction Cell Frontier: Analysis of Radionuclides in Environmental Samples using an Innovative Reaction Cell ICPICP-MSFP09Yasuyuki Shikamori, Kazumi Nakano, Naoki Sugiyama, and Shinichiro Kakuta, Agilent Technologies. Inc., 9-1 Takakura-cho, Hachioji-shi, Tokyo, 192-0033 JAPAN.Reference Materials and Calibration StandardsIntroductionLong-lived radionuclides have been released...
cps, cpsicp, icpiodine, iodinenuclear, nuclearisotopes, isotopesqqq, qqqchamber, chamberradionuclides, radionuclideslived, livedrsd, rsdradioactive, radioactiveabundance, abundanceqms, qmsaverage, averagebec