USP General Chapter <857> and Ph. Eur. Chapter 2.2.25 10th Edition
Others | 2019 | Agilent TechnologiesInstrumentation
Pharma & BiopharmaManufacturer
Agilent TechnologiesKey words
qualification, photometric, sodium, potassium, spectrophotometers, iodide, nitrate, didymium, vis, absorbance, test, services, acetone, accuracy, electronic, dichromate, lasts, cerium, wavelength, chloride, chapters, holmium, control, agilent, perchloric, stray, transmittance, xenon, offers, nicotinic, reproducibility, some, requirements, complies, shelf, records, pharmacopeia, flash, wavelengths, lamp, acid, quality, new, resolution, savings, manufactured, cases, oxide, nist, filters
Spectroscopy Solutions for Pharmaceuticals
2016|Agilent Technologies|Technical notes
Spectroscopy Solutions forPharmaceuticals:Confidence in Compliance to USP <857> Using theAgilent Cary 60 UV-Vis SpectrophotometerWhite PaperIntroductionUV-Vis spectroscopy is one of the most commonly used analytical techniquesin clinical chemistry, pharmaceutical research, and quality control/qualityassurance (QA/QC). Regulatory requirements for UV-Vis instruments used inthese...
winuv, winuvcary, carystray, straywavelength, wavelengthphotometric, photometricvalidate, validatespectrophotometer, spectrophotometerabsorbance, absorbanceaccuracy, accuracytests, testsvis, vislight, lightprecision, precisiontest, testwavelengths
Pharmaceutical Analysis Using UV-Vis: Compliance with USP Chapter <857>, and European Pharmacopoeia (Ph. Eur. Chapter 2.2.25)
Application NotePharmaceutical Analysis UsingUV-Vis: Compliance with USP Chapter<857>, and European Pharmacopoeia(Ph. Eur. Chapter 2.2.25)Meeting the requirements of the globalpharmacopeias with the Agilent Cary 3500 UV-VisIntroductionUV-Vis spectroscopy is a widely used analytical technique in quality assurance/quality control (QA/QC) and pharmaceutical research....
absorbance, absorbancewavelength, wavelengthphotometric, photometricstray, straylight, lightpositions, positionsholmium, holmiumvis, viscontrol, controltest, testreference, referencetolerances, tolerancestests, testsrecommended, recommendedspectrophotometer
Pharmaceutical Analysis using UV-Vis: Compliance with Supplement I to the Japanese Pharmacopoeia 18th Ed., Section 2.24
2021|Agilent Technologies|Technical notes
White PaperPharmaceutical Analysis usingUV-Vis: Compliance withSupplement I to the JapanesePharmacopoeia 18th Ed., Section 2.24Meeting the requirements of the globalpharmacopeias with the Cary 3500 UV-VisIntroductionUV-Vis spectroscopy is a widely used analytical technique in quality assurance/quality control (QA/QC) and pharmaceutical research. It...
wavelength, wavelengthphotometric, photometricvis, visoxide, oxideabsorbance, absorbancetests, testsdichromate, dichromateenter, enterspectrophotometer, spectrophotometerreference, referencepositions, positionsoperational, operationalprecision, precisionoptical, opticalaccuracy
Agilent Cary 3500 UV-Vis spectrophotometer series
2023|Agilent Technologies|Brochures and specifications
Multiply YourExperimental PossibilitiesAgilent Cary 3500 UV-Vis spectrophotometer seriesModular Design That SuitsYour Application NeedsThe Agilent Cary 3500 UV-Vis spectrophotometer series share a common UV-Visengine, which produces monochromatic light that is measured by the various UVVis sample measurement modules.This allows for the...
spectrophotometer, spectrophotometeropenlab, openlababsorbance, absorbancesecure, securecary, caryagilent, agilentvis, visdata, datastorage, storagecsv, csvspectrophotometers, spectrophotometerstemperature, temperatureesignatures, esignaturesworkstation, workstationyour