APWC: Arsenic(As) and Selenium(Se) analysis in presence of Rare Earth Elements (REEs) by single quadrupole ICP-MS
Posters | 2017 | Agilent Technologies Instrumentation
Environmental, Food & AgricultureManufacturer
Agilent TechnologiesKey words
matsue, sugiyama, naoki, rees, plasma, summary, singly, selenium, rare, ked, arsenic, kinetic, earth, ltd, discrimination, tuning, resolve, charged, having, affect, purpose, condition, elements, icp, japan, widely, helium, general, technologies, presence, unique, introduction, energy, spectra, cell, result, because, large, technique, quadrupole, three, conclusions, parameters, single, agilent, which, performance, ion, used, analysis
Handbook of ICP-QQQ Applications using the Agilent 8800 and 8900
5th EditionHandbook of ICP-QQQApplications using theAgilent 8800 and 8900Primer> Return to table of contents> Search entire documentForewordAgilent Technologies launched its 8800 Triple QuadrupoleICP-MS (ICP-QQQ) at the 2012 Winter Conference on PlasmaSpectrochemistry in Tucson, Arizona, USA.By the time the first ICP-QQQ...
return, returncontents, contentsicp, icptable, tableqqq, qqqcps, cpsgas, gasmass, massdocument, documentcell, cellppt, pptconc, concentire, entiresearch, searchelements
ICP-QQQ Applications in Geochemistry, Mineral Analysis, and Nuclear Science
Applications of ICP-MSICP-QQQ Applications inGeochemistry, Mineral Analysis,and Nuclear ScienceApplication Compendium> Return to table of contents> Search entire documentExplore New Opportunities in Geochemistry,Mineral Analysis, and Nuclear Science With theAgilent 8900 ICP-QQQAgilent atomic spectroscopy instruments are known for their robustness,excellent control of...
return, returncontents, contentstable, tableicp, icpree, reeqqq, qqqmass, masscps, cpscell, cellgas, gaselement, elementshift, shiftrare, rareybo, ybowwtp
Solving Doubly Charged Ion Interferences using an Agilent 8900 ICP-QQQ
2019|Agilent Technologies|Technical notes
Application NoteEnvironmentalSolving Doubly Charged IonInterferences using an Agilent8900 ICP-QQQImproved data quality using MS/MS oxygenmass-shift method to resolve common M++ion interferencesAuthorIntroductionNaoki Sugiyama,Agilent Technologies Inc.Most ICP-MS instruments use collision/reaction cell (CRC) technology to controlspectral interferences, greatly improving data accuracy for many elements...
doubly, doublyinterferences, interferencessrm, srmcharged, chargedicp, icpelements, elementsked, kedqqq, qqqion, iongas, gasaverage, averagecell, cellelement, elementcertified, certifiedrecovery
Resolving REE2+ Overlaps on Arsenic and Selenium With Hydrogen Cell Gas
2021|Agilent Technologies|Technical notes
Technical OverviewResolving REE2+ Overlaps on Arsenicand Selenium With Hydrogen Cell GasAccurate analysis of As and Se in a REE matrixusing Agilent single quadrupole ICP-MSIntroductionICP-MS is the technique of choice for rapid multi-element analysis of a widerange of trace and higher-level...
icp, icpked, kedcell, cellcorrection, correctionppb, ppbdoubly, doublyree, reepolyatomic, polyatomicbias, biascollision, collisioninterferences, interferenceshalf, halfgas, gaselements, elementsoverlaps